0.0001 to 3.0000 mL/min (up to 10.0 mL/min. at 220 bar)
Gradient Delay Volume
< 120 µL (equipped with 20 µL mixer)
Flow Rate Precision
Flow Rate Accuracy
2 Trays, Standard or deep 96-well & 105 x 1.5 mL vials
Room Temperature +5°C to 150°C
Injection Volume Range
0.1 to 50.0 µL
Injection Volume Precision
RSD ≤ 0.25% (5 µL injection)
< 14 sec.
Multiplate Autosampler An ideal replacement for the standard ExionLC AD Autosampler, the Multiplate Autosampler expands sample capacity over threefold without need for a separate Rack Changer. Three drawers can be configured to accept 6 regular or deep-well microtiterplates (MTP) in either 96 or 384-well formats. With the optional vial trays, the Multiplate Sampler can hold up to 324 x 1.5 mL vials.
Rack changer For maximum sample capacity, the ExionLC Rack Changer provides up to 12 additional cooled sample trays or vial racks. Each drawer can be configured for either regular or deep-well configurations, as well as 54-vial trays.
UV Detector A general purpose Variable Wavelength Detector, the ExionLC UV detector adds additional flexibility to your ExionLC AC or AD system. Time programmable wavelength switching allows detection sensitivity and selectivity for your specific analytical workflows.
PDA Detector The ExionLC PDA (PhotoDiodeArray) detector is an excellent option for identification, quantitation, and purity analyses at trace levels. A high sensitivity flow cell and low detector noise levels ensure uncompromised performance.
Solvent selector valve Need to change between gradient types ? The ExionLC Solvent Selector Valve allows a user to select between two solvents per channel from up to 3 different pumps.
Software has already been registered to another instrument. Please check your AID and try again. If you want to change the License Key to another instrument, please submit a new case.
Software has already been registered to the instrument selected. Please check your instrument's software by going to the Instruments page and selecting the instrument in your "My Instruments" section. Here you will see a Software tab where you can view your instrument's software information. You can also go to the Registered Software page to view software associated with your instrument(s). If you are not finding what you are looking for, please submit a new case.